List of discrete instrument responses of 14 common seismometers for 6 common sample intervals.

1 |

List of 14 lists (each corresponding to a different seismometer), each including 6 lists (each corresponding to a different sample interval), each containing the following elements:

- Sense
Instrument passband sensitivity (V * s/m)

- Knorm
Normalization constant

- poles
Poles of Laplace transform of instrument response (rad/s)

- np
Number of poles

- zeros
Zeros of Laplace transform of instrument response (rad/s)

- nz
Number of zeros

- dt
Sample interval for this response (s)

- fmax
Maximum frequency for which this digital response matches the true analog response of the sensor within 1%

- Zpg
Zpg-class element (from package 'signal') giving the digital response of the filter in terms of its zeros and poles (in Z-transform space) and gain.

Seismometers are numbered as follows:

Broadband Seismometers: 1. Streckeisen STS-1 (360 s) 2. Trillium 240 (generation 1) 3. Trillium 240 (generation 2) 4. Guralp CMG-3T 5. Streckeisen STS-2 (generation 1) 6. Streckeisen STS-2 (generation 2) 7. Streckeisen STS-2 (generation 3) 8. Trillium 120 9. Compact Trillium

Intermediate Seismometers: 10. Trillium 40 11. Guralp CMG-3ESP 12. Guralp CMG-40T (30 s) 13. Streckeisen STS-1 (20 s)

Short-period Seismometers: 14: Guralp CMG-40T (1 s)

Digital responses are provided for the following six sample intervals (in seconds): 1, 0.1, 0.05, 0.025, 0.02, 0.01.

The STS-2 and Trillium 240 come in multiple generations, each with a slightly different response. For the Trillium 240, serial numbers less than 400 belong to generation 1, while serial numbers greater than or equal to 400 are in generation 2. To determine which generation an STS-2 is, see http://www.iris.edu/NRL/sensors/streckeisen/streckeisen_sts2_sensors.htm .

Many short-period instruments were intentionally omitted because their responses depend on installation-specific parameters and are therefore not completely standardized. Given that discrete responses of short-period instruments can be calculated quickly, I consider the convenience of having pre-calculated responses for these instruments to not be worth the risk of the user selecting the wrong response and getting inaccurate results.

1 2 3 |

Questions? Problems? Suggestions? Tweet to @rdrrHQ or email at ian@mutexlabs.com.

Please suggest features or report bugs with the GitHub issue tracker.

All documentation is copyright its authors; we didn't write any of that.

Embedding an R snippet on your website

Add the following code to your website.

For more information on customizing the embed code, read Embedding Snippets.