Outputs clean rh_corrected_nephelometer measures for analysis.
the MicroPEM object
Boolean indicating whether there were measurements with HEPA filters at the beginning.
Boolean indicating whether there were measurements with HEPA filters at the end.
rh_corrected_nephelometer values are set to NA if they are negative or
if the RH at the same time is higher than 90%.
rh_corrected_nephelometer values are also corrected for the HEPA zeroings (start and end, if there were done):
if a stable period longer than 3 minutes can be identified for the HEPA period,
using the changepoint
cpt.mean function, there is a zero value.
There can be no zero values, only one (beginning or end) or two.
If there is only one zero value, it is substracted from all rh_corrected_nephelometer values.
If there are two, a linear interpolation is done between the two values and the resulting vector
is substracted from the rh_corrected_nephelometer values.
A MicroPEM object.
1 2 3
Add the following code to your website.
For more information on customizing the embed code, read Embedding Snippets.